Technical information: the indication range - 0-300mmHg, the measurement range - 60 / 300mmHg, the scale division - 2mmHg, the limits of the basic absolute error at an ambient temperature of 20 ± 5 ° C in mm Hg. in the measurement range from 60 to 240 mm Hg. - ± 3, in the measurement range from 240 to 300 mm Hg. - ± 4, additional error in measuring of the instrument at an ambient temperature different from (20 ± 5) ° С in the operating temperature range from + 5 ° С to + 45 ° С,% of the upper limit of measurements no more than ± kx (t20-t1 ), where: k - temperature coefficient equal to 0.06% ° C; t20 - ambient temperature equal to (20 ± 5) ° С; t1 - any temperature value corresponding to the operating conditions, Speed of air pressure reduction in the pneumatic system, mmHg. - 3 ± 2.
Instruction for use
It is intended for indirect determination of systolic and diastolicblood pressure by measuring the excess pressure in the cuff at the time of appearance and disappearance of Korotkoff's sounds
Store at temperature of -5 to + 40 ° C
over the counter
Original package - 24 pcs.
The information presented on the site should not be used for self-diagnosis and treatment and can not serve as a substitute for internal